Welcome back to “A Minute with Miranda.” This week we will be covering how the Ground Master Monitor provides continuous monitoring of the path-to-ground impedance and electromagnetic integrity of eight metal ground connections of process tools in your SMT assembly work area.
The Ground Master Monitor
continuously monitors eight metal tools for electromagnetic interference (EMI).
EMI can cause equipment lockups and malfunction. The Ground Master Monitor will
alarm if EMI is detected. The Ground Master will also alarm if the grounded
metal tools have a high-frequency noise that can cause electrical overstress
(EOS) damage. The Ground Master Monitor provides both a visual and audible
alarm for the monitored ground connections. The Ground Master Monitor meets the
Continuous Monitor requirements of ANSI/ESD S20.20 in accordance with ESD TR53.
With electronic components getting smaller and more sensitive, it’s important to make sure they are protected from ESD events like static discharge. Per ANSI/ESD S20.20, “Protective packaging is required to store, transport, and protect ESDS electronic items during all phases of production.” Per the new 2018 requirements for ANSI/ESD S541, the shielding requirement was changed that remaining discharge for the bags should be less than 20 nanojules.
One of the more common used bags is a low charging Pink Poly bag. These bags are made from a tinted polyethylene material with an antistatic coating that can wear away. This turns the bag insulative over time, making it noncompliant to ANSI/ESD S541 recommendations. They also lack discharge shielding protection which makes components within the bag susceptible to ESD event damage. Metallized Shielding bags are constructed from a metalized polyester film and a low charging polyethylene laminate. This provides the bags with a shielding layer that creates a Faraday cage protecting the ESD sensitive components within the bag from possible ESD event damage. The low charging inner layer and outer layer of the bag prevent tribocharging from occurring, minimizing the build up of ESD charges when handling components.
Watch this video on Pink Poly vs Static Shielding Bag Testing and learn why Metallized Static Shielding Bags are the best packaging solution offering full protection against ESD events.
Have you been questioned by your quality manager about shelf life on ESD bags? Maybe it’s an internal quality procedure that is being written and you would like to know where the manufacturer stands on the shelf life of ESD packaging. Well today we will address the topic of ESD Static Control bag storage and the expected shelf life.
There are many ways to manufacture an ESD shielding bag with low charging properties on the inner and outer layers. This is the first factor to consider so I will address only the items that I convert here at SCS. In our process, I achieve shielding values via a buried metal layer. The metal layer is laminated between two layers of polyethylene. This process makes a very robust and long- lasting product in terms of shielding values as per ANSI/ESD S541-2018, the electronic industry’s most recognized standard for ESD packaging. Still the metal layer is exposed to oxidization if not stored properly and subject to extreme climate or moisture.
To achieve the outer and inner low-charging properties, I use a topical anti-stat solution. This provides the outer and inner resistance readings of >/= 1 x 10^4 to < 1 x 10^11 ohms as called out in ANSI/ESD S541-2018 packaging standard. This property is the most vulnerable to change and should be tested frequently.
How does all this pertain to shelf-life you ask? SCS provides a Static Control Bag Storage document with a 1 year warranty from the date of purchase. I have witnessed SCS ESD bags stored in a controlled climate, well-ventilated area that have maintained passing surface resistance results after 5 yrs. on the shelf. Some customers have relied on our Moisture Barrier products to archive various critical products for as long as 20 yrs. They found through extensive testing that SCS had the highest quality and most repeatable results to trust with their archived product.
End users cannot always provide a climate and humidity-controlled atmosphere for our raw material, so it is recommended that you test your bags periodically per the S541 standard and dispose of any materials that are found to be questionable. As a best practice, you can log your products by purchase date or lot number and create an annual spot check audit to assure that your bags are still in compliance. Using a FIFO First In First Out process is critical in keeping your inventory fresh. You can imagine that there are 1000’s of variations to consider when we discuss shelf life. Heat, storage rack conditions, travel method and local climate to name a few, so the points to take away are this:
Hot, Cold, sunlight, moisture
Add controls when possible
In a controlled environment, SCS ESD bags have seen 5 yrs. of proven shelf life.
Follow the FIFO process as a best practice for inventory.
SCS warranties the product for a period of 1 year from the date of purchase. You need to audit annually or prior to use as a best practice and assure that your devices are getting the best protection.
If the bags are questionable, recycle the product at your local recycler and order and new lot from SCS – Static Control Solutions.
Thank you, Kevin Cleary ESD Packaging Specialist – Sales Direct Line: (919) 903-1724
Welcome back to “A Minute with Miranda.” This week we will be covering how the EM Aware Monitor provides continuous monitoring to detect and measure ESD Events in your STM machine.
The SCS EM Aware Monitor is a continuous monitor for three key parameters that allow you to verify your ESD process in an automated insertion machine; ESD events, change in static voltage field, and ionizer balance. The thresholds for all three of these parameters are fully adjustable by the user. The EM Aware Monitor is a miniature radio receiver tuned to detect and measure the unique waveform generated by an ESD event. The EM Aware Monitor meets the Continuous Monitor requirements of ANSI/ESD S20.20 in accordance with ESD TR1.0-01 and ANSI/ESD STM3.1. It meets the recommendations of ESD Handbook ESD TR20.20 which includes “if the products that are being produced are of such value that the knowledge of a continuous, reliable ground is needed, then continuous monitoring should be considered or even required.”
Welcome back to “A Minute with Miranda.” This week we will be covering how to use the WS Aware Monitor for continuous monitoring of SCS wrist straps when at an ESD Workstation.
The SCS WS Aware Monitor is a dual workstation continuous monitor for operators, ESD Worksurfaces and metal tools. It will continuously monitor the path-to-ground integrity and body voltage of two operators. The WS Aware Monitor eliminates the need for periodic testing and record keeping of wrist straps. The WS Aware Monitor features operator body voltage detection. It will alarm if the operator generates or comes into contact with voltage that would be dangerous to an ESD susceptible item. Per the ESD Handbook ESD TR20.20 section 18.104.22.168.4 “Typical test programs recommend that wrist straps that are used daily should be tested daily. However, if the products that are being produced are of such value that knowledge of continuous, reliable ground is needed, then continuous monitoring should be considered or even required.”
Welcome back to “A Minute with Miranda.” This week we will be covering how to properly clean an ESD Worksurface Mat.
For optimum electrical performance, the ESD worksurface mat surface should be cleaned regularly using a recommended ESD mat cleaner. Per the ESD Handbook ESD TR20.20, “Ensure that cleaners that are used do not leave an electrically insulative residue common with some household cleaners.” We recommend using Reztore ESD Surface and Mat Cleaner. Reztore is alcohol free and does not contain silicone or other substances that will leave an insulative residue or inhibit the performance of the ESD worksurface mats.
After cleaning the ESD worksurface mat with Reztore it is recommended to test the surface to ensure that all insulative contaminates such as dirt and grime have been removed from the mat. SCS worksurface mats meet the ANSI/ESD STM4.1 and ANSI/ESD S20.20 required limit of 1 x 106 to less than 1 x 109 ohms for Rtt and Rtg and the recommendations of ANSI/ESD S4.1. View the range of Reztore Alcohol Free products here.
Welcome back to “A Minute with Miranda.” This week we will be discussing how to test the point-to-point resistance (Rtt) and the resistance-to-ground (Rtg) of a Conductive ESD Floor.
ANSI/ESD S20.20 requires initial and periodic verification of an ESD Flooring System. ANSI/ESD STM7.1 outlines the test methods applicable for the Conductive flooring material. For the Point-to-Point resistance (Rtt) test the flooring will be tested with a resistance measurement meter and 2 x 5lbs cylindrical electrodes positioned 36” apart. The value for the test should be less than or equal to 1 x 106 ohms.
The Resistance Point-to-Ground (Rtg) test should be conducted with a resistance measurement meter and 1 x 5lbs cylindrical electrode. One lead from the meter should be connected to the ground point and the other lead will be connected to the electrode. The test value should be less than or equal to 1 x 106 ohms.
View the full range of SCS Surface Resistance Testers here.
Welcome back to “A Minute with Miranda.” This week we will be discussing how to test the point-to-point resistance (or Rpp) of an ESD Smock.
ANSI/ESD S20.20 requires initial and periodic verification of ESD Control items – this includes ESD Smocks. ANSI/ESD STM2.1 outlines the test method applicable for ESD Smocks: the ESD Smock is to be placed on an insulative surface and 2 x 5lbs cylindrical electrodes are to be positioned on each cuff before taking the measurement. The Resistance Point-to-Point Rpp of the groundable smock needs to be less than 1 x 109 ohms.
Welcome back to “A Minute with Miranda.” This week we will be covering how to properly wear a wrist strap.
ANSI/ESD S20.20 requires seated personnel to be
connected to the grounding / equipotential bonding system via a wrist strap.
The total resistance of the Wrist Strap System needs to be less than 3.5 x 10^7 ohms. The key to a wrist strap is
the intimate contact of the band to the skin and that the coil cord is
connected to ground. Wrist straps need to be tested at least daily before
handling any ESD sensitive devices.
Operators can choose between elastic and metal wristbands. Elastic wristbands are comfortable to wear and easy to adjust. Metal wristbands generally last longer and are easier to clean. View the full range of SCS Wrist Straps here.
Welcome back to “A Minute with Miranda.” This week we will be covering how to perform the Rtt and Rtg test on an ESD Worksurface Mat.
Before using a worksurface mat in an EPA environment, you need to verify that the mat will meet the ANSI/ESD S20.20 Worksurface Requirements. For both the Rtt and Rtg test the worksurface should test between 1 x 106 to less than 1 x 109 ohms. The test should be performed in accordance to the ANSI/ESD S4.1 Standards.
SCS worksurface mats meet the ANSI/ESD STM4.1 and ANSI/ESD S20.20 required limit of 1 x 10^6 to less than 1 x 10^9 ohms for Rtt and Rtg and the recommendations of ANSI/ESD S4.1.
View our complete SCS Static Worksurface mat offering here.