Welcome back to “A Minute with Miranda.” This week we will be discussing how to test the point-to-point resistance (or Rpp) of an ESD Smock.
ANSI/ESD S20.20 requires initial and periodic verification of ESD Control items – this includes ESD Smocks. ANSI/ESD STM2.1 outlines the test method applicable for ESD Smocks: the ESD Smock is to be placed on an insulative surface and 2 x 5lbs cylindrical electrodes are to be positioned on each cuff before taking the measurement. The Resistance Point-to-Point Rpp of the groundable smock needs to be less than 1 x 109 ohms.
Welcome back to “A Minute with Miranda.” This week we will be covering how to properly wear a wrist strap.
ANSI/ESD S20.20 requires seated personnel to be
connected to the grounding / equipotential bonding system via a wrist strap.
The total resistance of the Wrist Strap System needs to be less than 3.5 x 10^7 ohms. The key to a wrist strap is
the intimate contact of the band to the skin and that the coil cord is
connected to ground. Wrist straps need to be tested at least daily before
handling any ESD sensitive devices.
Operators can choose between elastic and metal wristbands. Elastic wristbands are comfortable to wear and easy to adjust. Metal wristbands generally last longer and are easier to clean. View the full range of SCS Wrist Straps here.