Welcome back to “A Minute with Miranda.” This week we will be covering how to use the WS Aware Monitor for continuous monitoring of SCS wrist straps when at an ESD Workstation.
The SCS WS Aware Monitor is a dual workstation continuous monitor for operators, ESD Worksurfaces and metal tools. It will continuously monitor the path-to-ground integrity and body voltage of two operators. The WS Aware Monitor eliminates the need for periodic testing and record keeping of wrist straps. The WS Aware Monitor features operator body voltage detection. It will alarm if the operator generates or comes into contact with voltage that would be dangerous to an ESD susceptible item. Per the ESD Handbook ESD TR20.20 section 18.104.22.168.4 “Typical test programs recommend that wrist straps that are used daily should be tested daily. However, if the products that are being produced are of such value that knowledge of continuous, reliable ground is needed, then continuous monitoring should be considered or even required.”
Welcome back to “A Minute with Miranda.” This week we will be covering how to properly clean an ESD Worksurface Mat.
For optimum electrical performance, the ESD worksurface mat surface should be cleaned regularly using a recommended ESD mat cleaner. Per the ESD Handbook ESD TR20.20, “Ensure that cleaners that are used do not leave an electrically insulative residue common with some household cleaners.” We recommend using Reztore ESD Surface and Mat Cleaner. Reztore is alcohol free and does not contain silicone or other substances that will leave an insulative residue or inhibit the performance of the ESD worksurface mats.
After cleaning the ESD worksurface mat with Reztore it is recommended to test the surface to ensure that all insulative contaminates such as dirt and grime have been removed from the mat. SCS worksurface mats meet the ANSI/ESD STM4.1 and ANSI/ESD S20.20 required limit of 1 x 106 to less than 1 x 109 ohms for Rtt and Rtg and the recommendations of ANSI/ESD S4.1. View the range of Reztore Alcohol Free products here.
Welcome back to “A Minute with Miranda.” This week we will be discussing how to store and ship your static sensitive assemblies within and outside of an ESD Protected Area using an In-Plant Handler.
ANSI/ESD S20.20 requires that ESD protective packaging is necessary to store, transport and protect ESD sensitive electronic items during all phases of production and shipment. Beyond the static protection, packaging also provides protection from physical damage, moisture, dust and other contaminates. Per the ANSI/ESD S541 Packaging Standard packaging that is used inside and outside of an EPA shall be low charge generating, constructed from dissipative or conductive material for intimate contact with ESD items and provide electrostatic discharge shielding.
The Protektive Pak In-Plant Handlers are constructed from an impregnated dissipative corrugated material that has a surface resistance range of 1 x 106 to less than 1 x 109 ohms per ANSI/ESD STM11.11. This minimizes the potential of rapid discharge or sparking. The in-plant handlers will shield ESD sensitive items from charge and electrostatic discharges with the lid in place. The partition sets are constructed from the same impregnated dissipative material and are available in over 300 cell size configurations to suit your specific packaging needs.
View the full range of Protektive Pak In-Plant Handlers here.
Welcome back to “A Minute with Miranda.” This week we will be discussing how to test the point-to-point resistance (Rtt) and the resistance-to-ground (Rtg) of a Conductive ESD Floor.
ANSI/ESD S20.20 requires initial and periodic verification of an ESD Flooring System. ANSI/ESD STM7.1 outlines the test methods applicable for the Conductive flooring material. For the Point-to-Point resistance (Rtt) test the flooring will be tested with a resistance measurement meter and 2 x 5lbs cylindrical electrodes positioned 36” apart. The value for the test should be less than or equal to 1 x 106 ohms.
The Resistance Point-to-Ground (Rtg) test should be conducted with a resistance measurement meter and 1 x 5lbs cylindrical electrode. One lead from the meter should be connected to the ground point and the other lead will be connected to the electrode. The test value should be less than or equal to 1 x 106 ohms.
View the full range of SCS Surface Resistance Testers here.
Welcome back to “A Minute with Miranda.” This week we will be discussing how to test the point-to-point resistance (or Rpp) of an ESD Smock.
ANSI/ESD S20.20 requires initial and periodic verification of ESD Control items – this includes ESD Smocks. ANSI/ESD STM2.1 outlines the test method applicable for ESD Smocks: the ESD Smock is to be placed on an insulative surface and 2 x 5lbs cylindrical electrodes are to be positioned on each cuff before taking the measurement. The Resistance Point-to-Point Rpp of the groundable smock needs to be less than 1 x 109 ohms.
Welcome back to “A Minute with Miranda.” This week we will be covering how to properly wear a wrist strap.
ANSI/ESD S20.20 requires seated personnel to be
connected to the grounding / equipotential bonding system via a wrist strap.
The total resistance of the Wrist Strap System needs to be less than 3.5 x 10^7 ohms. The key to a wrist strap is
the intimate contact of the band to the skin and that the coil cord is
connected to ground. Wrist straps need to be tested at least daily before
handling any ESD sensitive devices.
Operators can choose between elastic and metal wristbands. Elastic wristbands are comfortable to wear and easy to adjust. Metal wristbands generally last longer and are easier to clean. View the full range of SCS Wrist Straps here.
Welcome back to “A Minute with Miranda.” This week we will be covering how to perform the Rtt and Rtg test on an ESD Worksurface Mat.
Before using a worksurface mat in an EPA environment, you need to verify that the mat will meet the ANSI/ESD S20.20 Worksurface Requirements. For both the Rtt and Rtg test the worksurface should test between 1 x 106 to less than 1 x 109 ohms. The test should be performed in accordance to the ANSI/ESD S4.1 Standards.
SCS worksurface mats meet the ANSI/ESD STM4.1 and ANSI/ESD S20.20 required limit of 1 x 10^6 to less than 1 x 10^9 ohms for Rtt and Rtg and the recommendations of ANSI/ESD S4.1.
View our complete SCS Static Worksurface mat offering here.
Welcome back to “A Minute with Miranda.” This week we will be covering how to launder your ESD Smocks.
SCS Static Control Smocks should be laundered periodically for proper operation. Smocks should be washed by hand or with a standard household washing machine using cold or warm water with a non-ionic liquid detergent. Avoid using bleach, fabric softeners or dry detergents as these can contaminate the conductive fibers and make the garment insulative .
After washing the smocks they should either be hung dry or tumbled dry at low heat. High heat can degrade the conductive fibers within the garment and degrade the ESD properties.
With normal wearing and washing conditions, SCS Static Control Smocks will maintain their ESD properties for a minimum of 100 washings.
SCS Smocks meet the requirement for Groundable Static Control Garment System per ANSI/ESD S20.20 required limit of less than 3.5 x 107 ohm Rtg tested per ANSI/ESD STM2.1 and ESD TR53.
SCS is excited to announce a brand new video series discussing real-life ESD problems and solutions. This is a great educational resource for anybody new to ESD or just wanting to learn more about best practices.
Each episode will focus on one issue commonly found in an ESD Protected Area – at the same time we will present solutions so you know how to tackle the problem should you ever face it in your own factory.
A new episode will be published each week so make sure you subscribe to our YouTube channel to get notified when a new video is available. Episodes 1 and 2 are now live so don’t waste a second longer and catch-up now: